B&K Function & RF Generators
2 MHz Sweep/Function Generator
Model 4017 * LIN/LOG sweep mode operation * TTL pulse & CMOS pulse outputs * Covers 0.2 Hz to 2.0 MHz in 7 ranges* Voltage Controlled Frequency (VCF) Input Amplitude output, plus 20 dB step attenuator Part No. 3T3-517 $369.00 |
SPECIFICATIONS (All specs apply with frequency dial between 0.2 and 2 times range.) Basic Outputs: Sine, square, triangle, TTL pulse, CMOS pulse. Freq. Dial Accuracy: +-5%. Max. Amplitude: 20V p-p (open circuit). 10V p-p (into 50 ohm load). Amplitude Control: Continuously variable, 30dB range typical. Attenuator: -20dB +-1dB. Output Impedance: 50 ohm +-6%. DC Offset: Continuously variable, from -10V to +10V (open circuit), -5V to +5V into 50 ohm. Duty Cycle Control: Continuously variable from 1:1 to 10:1. Sine Wave Distortion: 0.2Hz to 200kHz. Square Wave Symmetry 0.2Hz to 100kHz. Triangle Wave Linearity: 0.2Hz to 100kHz, 98%. CMOS Output: Level: Continuously adj. from 4V p-p (+-1V p-p) to 14.5V p-p (+-0.5V p-p). VCF (Voltage Controlled Freq.) Input: Voltage: Approx. +10V (+-1V) causes 1000:1 freq. change. Impedance: 10k ohms +-5%. Sweep Operation: Auto/Manual: Switch selectable. Sweep Width: Continuously adj. General Specs: Power: 120/220/240 VAC +-10%, 50/60Hz. Dimensions (HxWxD): 9.33 x 3.33 x 11.8" (237 x 85 x 284mm). Weight: 6.6 lbs. (3.0kg). Accessories Included: One cable, BNC to insulated clips, power cord, spare fuse, instruction manual, schematic diagram and parts list. |
2 MHz Function Generator with Digital Display
Model 3011B * Variable DC offset, w/ON/OFF switch * Variable duty cycle; CAL position for 50% duty cycle * Variable amplitude output, plus 20 dB step attenuator * Coarse and fine frequency controls * Generates sine, square, triangle, pulse & ramp waveforms * TTL pulse & CMOS pulse outputs Part No. 3T3-511 $284.00 |
SPECIFICATIONS (All specs apply with frequency dial between 0.2 and 2 times range.) Basic Outputs: Sine, square, triangle, TTL pulse, CMOS pulse. Freq. Range: 0.2Hz to 2MHz (7 ranges). Four digit freq. counter display. Max. Amplitude: 20V p-p (open circuit). 10V p-p (into 50 ohm load). Amplitude Control: Continuously variable, 20dB range typical. Output Impedance: 50 ohm +-6%. Duty Cycle Control: Continuously variable from 1:1 to 10:1. Sine Wave Distortion: 0.2Hz to 20kHz less than 1%; 20kHz to 200kHz less than 2%. Square Wave Symmetry: 0.2Hz to 100kHz less than 2%. Triangle Wave Linearity: 0.2Hz to 100kHz - 98%; 100kHz to 2MHz - 95%. CMOS Output Level: Continuously adjustable from 4V p-p to 14.54V p-p. VCF Input: Input Voltage approx. +10V causes 10:1 freq. change. Impedance: 10k ohms. Freq. Counter (Internal Only): Accuracy: +- Time Base. Accuracy: +- 1 count. Time Base Accuracy: +- 10 PPM. General Specs: Power: 120/220/240 VAC, 50/60Hz. Dimensions (HxWxD): 3.33 x 9.33 x 11.18" (85 x 237 x 284mm). Weight: 5.5 lbs. Accessories Included: One cable, BNC to insulated clips, power cord, spare fuse, instruction manual, schematic diagrams and parts list. |
20 Hz-150 kHz Sine/Square Wave Audio Generator
Model 3001 * Low battery indication * Variable output control; * 20 Hz to 150 kHz in 46 steps * Low distortion R-C oscillator * Lightweight, portable design * Sine & square wave generator Part No. 3T3-501 $79.00 |
SPECIFICATIONS Frequency Characteristics: Range: x1 range 20Hz to 1.5kHz (23 steps), x100 range 2kHz to 150kHz (23 steps). Accuracy: 20Hz through 100kHz (3% or less), 120kHz and 150kHz (5% or less). Output Control: 0dB-20dB attenuator switch and variable amplitude control. Output Impedance: 600 ohms. Sinewave Characteristics: Output Voltage 1.2V rms at max. setting (no load). Output Flatness (Short Term): 20Hz to 150kHz (0.5dB). Square Wave Characteristics: Output Voltage: 5V p-p at max. setting. Duty Ratio: 50%. Synchronization Characteristics: Output Voltage: 1.2V rms (no load). Output Impedance: 1k ohms. Other specs same as sine wave. General Specs: Power: 9V battery. Battery Life: 35 hours, alkaline. Battery Indicator: LED indicates low battery. Dimensions (HxWxD): 6 x 3.3 x 0.9" (15 x 8.2 x 2.1cm). Weight: 7 oz. (200g) including battery. Accessories Included: Two standard banana plug to insulated clip test leads, 9V battery, instruction manual, schematic diagrams and parts list. |
150 MHz RF Signal Generator
MODEL 2005 * Output to 450 MHz on harmonics * 100 kHz to 150 MHz on six bands * Step and variable attenuation to -40dB * AM modulation, int. or ext., 0% - 100% * Frequency monitor output for ext. frequency counter Part No. 3T4-505 $275.00 |
SPECIFICATIONS MAIN OUTPUT Frequency Range: 100kHz to 150MHz (up to 450MHz on harmonics). A) 100kHz-300kHz. B) 300kHz-1000kHz. C) 1.0MHz-35MHz. D) 3.2MHz-11.0MHz. E) 10.0MHz-3.50MHz. F) 32MHz-150MHz (96-450MHz on harmonics). Dial Accuracy: +-3%. RF Output Level: Approx. 100mV RMS max. (up to 35MHz). Output Level Control: Hi-Lo step attenuator approx. 20dB. Variable in Hi or Lo step, at least 20dB range of adjustment. FREQUENCY MONITOR OUTPUT Frequency: 100kHz to 150MHz, same as main output. Level: 30mV RMS typical, fixed. AMPLITUDE MODULATION Internal: Frequency 1kHz. Level Continuously Variable 0 to 100% percentage of modulation. External: Frequency: 50Hz to 20kHz. Sensitivity: less than 1V RMS. AUX OUTPUT Audio Output: Freq. 1kHz Level: 1V RMS min. (fixed). GENERAL Power Requirements: 120 VAC +-10%, 60Hz. Dim. (HxWxD): 158 x 247 x 140mm. Typical Wt: 2.5Kg. BNC to Insulated Clip Output Cable, Detachable Power Cord, Manual. |
Miscellaneous
3T3-522 |
3022 |
2 MHZ GENERATOR W/FREQ COUNTER |
Was 479.00 now $365.00 |
3T3-540 |
4040 |
13 MHZ GENERATOR W/FREQ COUNTER |
$499.00 |
Last Updated April 24, 2000